- Cancer Biology & Biomarkers
- Chromatography & Mass Spectrometry
- Contract Research, Clinical Trials and Outsourcing
- Drug Discovery
- Drug Targets
- Flow Cytometry
- Informatics & Lab Automation
- Ingredients, Excipients and Dosages
- Microbiology & RMMs
- NIR, PAT & QbD
- Raman Spectroscopy
- Screening, Assays & High-Content Analysis
- Thermal Processing
- Events & Workshops
Whitepaper: Changes to GMP force cleanroom re-classifications
3 March 2017 • Author(s): Beckman Coulter
According to both USA cGMP1 and EU GMP2 cleanroom classification should be carried out according to ISO 14644-1.
The 19994 version of this ISO standard has been re-written by a group of international subject matter experts and ISO 14644-1:20156 was published on December 15, 2015. ISO 14644-1:2015 contains substantial revisions impacting both the way the cleanrooms are classified and the performance requirements of the air particle counting instruments used to carry out the classification. Cleanroom users claiming compliance to cGMP and/or GMP will have to consider these new rules when next re-qualifying their cleanrooms, potentially purchasing new air particle counters if their existing counters do not comply with the new performance rules and facing the dilemma that the new re-qualification may indicate that their cleanroom is not operating at the grade they had achieved using the old classification process. This paper explains the new rules.
Although EU GMP and USA cGMP both define the airborne particle counts/m3 for each cleanliness class/grade, they do not contain guidance on the cleanroom classification procedure. Instead they point the reader to ISO 14644-1 for the cleanroom classification methodology. As part of the routine periodic review process, the ISO committee responsible for ISO 14644-1 decided that areas of the classification process required revision to improve statistical accuracy. At the same time, the ISO committee decided to add a reference in the normative Annex on airborne particle counters to ISO 21501-45, which provides a calibration procedure and verification method for particle counters, intended to minimize the inaccuracy in the measurement result by a counter, as well as the differences in the measured results from different particle counters. The stringent performance rules in ISO 21501-4 mean that many older air particle counters are no-longer suitable for cleanroom classification/re-qualification according to ISO 14644-1.
ABB Analytical Measurement ACD/Labs ADInstruments Ltd Advanced Analytical Technologies GmbH Analytik Jena AG Astell Scientific Ltd B&W Tek Bachem AG Bibby Scientific Limited Bio-Rad Laboratories BioNavis Ltd Biopharma Group Black Swan Analysis Limited CAPSUGEL NV Charles Ischi AG | Kraemer Elektronik Cherwell Laboratories CI Precision Cobalt Light Systems Coulter Partners CPC Biotech srl Dassault Systèmes BIOVIA DiscoverX Edinburgh Instruments Enterprise System Partners (ESP) Eurofins BioPharma Product Testing EUROGENTEC F.P.S. Food and Pharma Systems Srl GE Analytical Instruments IDBS JEOL Europe L.B. Bohle Maschinen + Verfahren GmbH Lab M Ltd. LabWare Linkam Scientific Instruments Limited MA Business Metrohm Molins Technologies Multicore Dynamics Ltd Nanosurf New England Biolabs, Inc. Ocean Optics Panasonic Biomedical Sales Europe B.V. PerkinElmer Inc ReAgent Russell Finex Limited Source BioScience Takara Clontech Tornado Spectral Systems Tuttnauer Viavi Solutions, Inc Watson-Marlow Fluid Technology Group Wickham Laboratories Limited Xylem Analytics YMC Europe GmbH Yusen Logistics