- Cancer Biology & Biomarkers
- Chromatography & Mass Spectrometry
- Contract Research, Clinical Trials and Outsourcing
- Drug Discovery
- Drug Targets
- Flow Cytometry
- Informatics & Lab Automation
- Ingredients, Excipients and Dosages
- Microbiology & RMMs
- NIR, PAT & QbD
- Raman Spectroscopy
- Screening, Assays & High-Content Analysis
- Thermal Processing
- Events & Workshops
Universal detection in high performance liquid chromatography
3 July 2015 • Author(s): David P. Elder, Phil Borman and George Okafo - GlaxoSmithKline
The most commonly used analytical technique for the testing of drug molecules is reversed-phase high performance liquid chromatography (RP-HPLC) with ultraviolet (UV)/visible detection. However, one of the biggest constraints affecting UV/visible detectors is their inability to generate adequate sensitivity for analytes lacking a suitable UV chromophore. These compounds comprise a surprisingly large group, encompassing natural products, particularly fermentation products, carbohydrates, lipids, certain amino acids, steroids, bisphosphonates, alkylating agents, excipients, polymer, surfactants, etc. (see Figure 1, page 50, for some representative structures). Secondly, it is often assumed that response factors for impurities are the same as for the parent when using UV detection systems, when often they are not1. Accordingly, purity assessments are inevitably compromised to some degree. This short review will look at the use of some newer detection systems and the pros and cons of the various detector types. Hyphenated techniques such as HPLC-NMR and HPLC-MS have been deliberately excluded.
Historically, refractive index (RI), indirect UV/visible or post-column derivatisation (followed by UV/visible detection) and conductivity have all been employed to address the deficiencies in HPLC analysis. In parallel, gas chromatography/flame ionisation detection (GC/FID) has been used as a universal detector for the more volatile analytes. However, these alternative detection approaches also have intrinsic limitations.
Newer detectors for non-chromophoric analytes that have been developed over the past few decades have entered into reasonably common usage. These include the aerosol-based detection systems, e.g., evaporative light scattering detection (ELSD), the closely related nano quantity analyte detector (NQAD), sometimes referred to as condensation nucleation light-scattering detection (CNSLD), corona-charged aerosol detection (CAD), and chemiluminescent nitrogen-specific detection (CLND).
ABB Analytical Measurement ACD/Labs ADInstruments Ltd Advanced Analytical Technologies GmbH Analytik Jena AG Astell Scientific Ltd B&W Tek Bachem AG Bibby Scientific Limited Bio-Rad Laboratories BioNavis Ltd Biopharma Group Black Swan Analysis Limited Butterworth Laboratories Ltd CAPSUGEL NV Charles Ischi AG | Kraemer Elektronik Cherwell Laboratories CI Precision Cobalt Light Systems Coulter Partners CPC Biotech srl Dassault Systèmes BIOVIA DiscoverX Edinburgh Instruments Enterprise System Partners (ESP) Eurofins BioPharma Product Testing EUROGENTEC F.P.S. Food and Pharma Systems Srl GE Analytical Instruments IDBS JEOL Europe Kaiser Optical Systems Inc. L.B. Bohle Maschinen + Verfahren GmbH Lab M Ltd. LabWare Linkam Scientific Instruments Limited MA Business Metrohm Molins Technologies Multicore Dynamics Ltd Nanosurf New England Biolabs, Inc. Ocean Optics Panasonic Biomedical Sales Europe B.V. PerkinElmer Inc ReAgent Russell Finex Limited Source BioScience Takara Clontech Tornado Spectral Systems Tuttnauer Viavi Solutions, Inc Watson-Marlow Fluid Technology Group Wickham Laboratories Limited Xylem Analytics YMC Europe GmbH Yusen Logistics